Comment on page
Troubleshooting MiSeq "Z Motor The Z stage amp Op amp might not be wired correctly" Error
Users may encounter a "Z stage/amp/Op-amp might not be wired correctly" error at any point during a MiSeq run. This issue may be caused by a communication error or by issues with the Z-stage itself.
To troubleshoot this issue :
- 1.Perform the post-run wash as indicated.
- 4.If error persists, instrument fails to initialize, or one of the Z-Stage test replicates fail, Contact Illumina Technical Support.
- 5.If the instrument initializes without error and all five Z-Stage tests pass, the instrument is clear for the next run.