PhiX loading concentrations for validation runs on Illumina sequencing platforms
A PhiX validation run confirms proper hardware and software performance of the instrument. The Illumina PhiX control library is a well-balanced genome with relatively equal representation of A, T, G, and C nucleotides. PhiX lacks an index and is not an appropriate tool for assessing Index Read performance.
It is important that a PhiX control run fall within optimal cluster densities for each platform, as listed in the following table.
Optimal Loading Concentration
Optimal Raw Cluster Density
MiSeq v2 reagents
MiSeq v3 reagents
NextSeq 500/550 High Output reagents
NextSeq 500/550 Mid Output reagents
HiSeq 2000/2500 High Output v3
HiSeq 2000/2500 High Output v4
HiSeq 2500 Rapid Run v2
NovaSeq - Standard Workflow
NovaSeq - XP Workflow
*Nondenatured pre-ExAmp concentration **Patterned flow cells consist of a nanowell with ordered wells, no variation in reported cluster density from run to run ***Final loading concentration
Run length and parameters vary depending on troubleshooting needs, however, Read 1 and 2 must have at least 26 cycles to generate quality scores.
For further information about preparing PhiX for sequencing, see the following documents.
For any feedback or questions regarding this article (Illumina Knowledge Article #1536), contact Illumina Technical Support [email protected].