Comment on page
Troubleshooting Cavro Pump Error on MiSeq
A Cavro Pump error can result from an issue with the instrument hardware or communication errors on the MiSeq. Potential hardware related causes include pump, pump controller, or syringe failure.
Error: Cavro pump error: Volume will exceed syringe capcity.Error in LanePump. Invalid argument. Command = '1V400IP1500R\r', reply = 'c'
Troubleshooting Steps:1. Power cycle* the MiSeq. (Note: If a run is in progress, it will not be able to be resumed.) 2. If there are no issues with re-initialization:
- Perform post-run wash to ensure fluidics is operating normally.
- Optionally, perform a volume test within the system check.
For further information, search for the following Knowledge Base articles:* *How to Shut down or power cycle the MiSeq.
- *How To Power Cycle the MiSeq Video.
- System Checks for MiSeq.
- Troubleshooting Lane Pump Errors on the MiSeq.