Comment on page
Troubleshooting a Failed to Get Temperature error on the MiSeq
The following warning is typically a transient communication issue between the MiSeq Control Software and a temperature sensor and does not affect run performance.
"@@@---Failed to get temperature."
- 1.If run performance is normal, power cycle instrument the instrument after the run completes and continue with the next run if the system initializes without issue.
- 2.If run performance is affected, power cycle instrument and perform system check tests. If all tests pass, then proceed with next sequencing run.
For reference, search Knowledge Base articles:
- How to Shut Down or Power Cycle the MiSeq
- How to perform a MiSeq system check
Last modified 12d ago